Load Monitor Analyzer (LMA) is a comprehensive solution for measuring & testing of push-in strength and pull out strength of an object. It is very useful tools which record online data and store it for future analyses. It has an interface to view old data and even export & printing of captured data. It has software and a microcontroller-based smart device. Which collects analog or digital signals, process them and transfer it to a computer. It is fast, accurate and easy to operate.
The entire system has two physical parts:
• LOAD MONITOR
This manual is applicable only for software installation, setup, and operations.
SOFTWARE: Software is an integral of the LMA or you can say software is the brain of the entire system. This will not work without software. So, it is important to understand the software.
The software manual is organized in the following sections:
• Installation of the software
• Configuration of the software
• Operation on the software
• Export of software data to other software.
The process of loading the software on the computer is called software installation. Before installing the software it is important to note that the computer has the following minimum requirements:
1. Computer with Pentium 4 processor, 1 GB RAM, 500 MB free space for software installation and 1 GB Free hard disk space for software running.
2. The software is perfectly tested on windows 2000 Profession and XP with service pack 2.0 and 3.0. Hence it is observed that it can be installed on Windows 95 and other flavors of windows yes you need following extra components to be loaded on those computers:
a. MDAC 2.8
b. DCOM for Windows 95/98
c. Above software are free and can be downloaded from www.microsoft.com
3. One DB9 RS232 Serial port free to connect the device.
Loading the software on the Computer is very easy. Just insert the CD into the CD Drive and locate either Setup.exe or Setup.MSI and double click on it. Just provide the information asked by the installation wizard and software is ready. Note: - if you don’t understand just press the next button on the screen would install the software to the default location on your computer. Followings are the default location:
1. “C:\program files\Loadana” is the path where your software is installed.
2. “C:\program files\Loadana\data” is the path where your data will be saved.
3. “C:\programfiles\Loadana\Server.cfg” is the main file where all software settings are stored.
4. “C:\program files\ Loadana\data\data.mdb” is the default data file.
5. You can open the “C:\program files\FMS\Server.cfg” using notepad windows standard program to update the data settings.
a. You will find DATA=data\data.mdb line by default.
b. You can change this to your desired location and file name.
c. Just make sure that the path you have changed exists and the data.mdb file is copied and renamed with a name you have given.
Just click on the start button and you will find on the FMS Icon. On click it software start running. If your computer does not have com1 it will give an error that “Com1 is not found on the computer” and you will see the following the main window of the software.
DEVICE SETTING PASSWORD SCHEME
Device Setup password = day X 2 month X 3 i.e. say today’s date id 08/08/08 then PWD is 1624 and 0n next day PWD is 1824.
SETTING DEVICE SETTINGS
These device settings are controlling the behavior of the Load Monitor device.
Com Port = number (1-99) DB9 connection number on the computer.
Pol delay = Milliseconds max value 31000. Request for data from the device after the given interval.
Re-Try Delay = Milliseconds max value 31000. In case no response from the device of a given command then send the same command again.
Next CMD Delay = Milliseconds max value 31000. The delay between successful data receiving from one command to the next command.
Max Tries = Max no(s) of tries software should be made to a device before declaring it communication failure.
Multiplication factor = Is the Zoom size of a given value of load.
Decimal Value DP1 = Dot point position selection in the red display.
Decimal Value DP2 = Dot point position selection in a green display.
Load Unit = Select your load Unit.
Elongation Unit = select elongation unit.
Threshold value = Minimum value after which data will be treated as valid data and automatically new test start.
Calibration Value = is the load value by which the load cell is calibrated. This command only effective if you press the calibrate button and receive a successful reply.
Set Decimal points button – is needed to be pressed if you wish to change the display format changed on DP1 & DP2 display immediately.
Set values are the values on which load monitor would ignite the relay on reaching the load value to set values. Pass/Fail result is shown if load values are between setval1 and setval2.
This option is taking input of the test no to display the graph on the screen. In case of no valid test nothing will happen.
MAIN SCREEN SETTINGS
There are certain settings provided on the screen, which required on the fly when performing the test. It is important to understand these settings:
Material Type – would be printed on the report.
Material Size– would be printed on the report.
Unit– would be printed on the report.
Shape– would be printed on the report.
Max Load – required for setting the Maximum value of the graph.
Elongation - required for setting the Maximum value of the graph.
Time Sec - required for setting the Maximum value of the graph.
Test Name - – would be printed on the report.
SET – clicking on the set button would save and set the options.
RESETTING ELONGATION VALUE TO ZERO
Pressing the light red button captioned as RESET ELONGATION would reset the elongation.
PRINTING REPORT & GRAPH
Report and graphs can be printed by clicking on the print button on the main screen. When clicking on the print button you will get two options.
Print Test Report
Clicking on the print test report option would ask you to enter test no. input a valid test no would print open the report in a new window. There may chances that you will not get
proper formatting of the report. To resolve this problem just do the following:
Click on the Format->font Window. Select Font Name = “Courier New” press OK.
Printing can be done by selecting the print command or just press Ctrl+P to print the report.
Printing Graph can be done by selecting the print graph option. Input the test no and select a graph to print. You have the option to view the graph before print.
Data storage is shown on the main screen on the upper left hand of the screen. You can change the storage location and file to the location of your preference. To change the data storage location, you have to click on the yellow strip of the current data storage location. Say you want to set data location to d:\loadmonitor\data_jan2008 then click on the current data location would open the browse location window, you have to locate the above location and give a file name of your choice would create the new data file and data storage location.
You can Export data by clicking on the Export button on the main window. You have the option to export report data to text format or excel file format.
INTEGRATION WITH ERP
If you have any software you standard ERP system you can integrate this software directly to your system using its COM-based SDK (Not free).
Paid telephonic service can be obtained by 9:00-17:00 IST for the software. Email at email@example.com
- Stock: In Stock
- Model: LOAD MONITOR ANALYZER
- Weight: 0.00kg